Moreover I attach some printscreens where it can be seen how I defined the channel and source. Message 10 of Source is included so that the driver can be modified if desired. Message 5 of 6. If you do not have internet access, the drivers can be obtained by contacting your country support organization. I looked on the manual for this error and I cound not find it. I have done many kinds of sweeping on this one, but this is my first time with sampling and I can’t seem to figure that one out.
|Date Added:||24 January 2018|
|File Size:||40.20 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Do not build the driver in the directory in which it was installed. All of the directory paths which follow assume Windows The SCPI commands look similar for both, but there are probably subtle differences that are the result 1456b new features in the Agilent C.
Agilent Analyzer – IEEE (GPIB) Driver for LabVIEW – National Instruments
Its actually a pretty poor example, I’m surprised its in the driver package like that. Norbert Norbert ,abview Source is included so that the driver can be modified if desired. Hey again, This is providing the source voltages and also measuring through the SMUs perfectly.
I will keep you posted with the outcome. Message 5 of 6.
Agilent 4156 Analyzer
The help file presents application programming examples, a cross-reference between instrument commands and driver functions, and detailed documentation of each function, with examples.
Most Active Software Boards: Thank you for all the support. I really appreciate all the feedback I can receive. The data I want to represent is IV curves. I have done many kinds of sweeping on this one, but this is my first time with sampling and I can’t labvview to figure that one out.
After this I can see the cursor on the bottom of the I V graph on the monitor of the sliding from left 41556b right when settinp a Sweep of Voltage of SMU1 from 0 to 1 V, pts. Message 9 of You can right click on the error cluster which shows error 6 and select “Explain Error”.
Hello, it’s me again with my very old and troubled semiconductor analyzer.
Second, there is no knowledge base file, which is primarily a description of a VXI board. Most Active Software Boards: After detailing a bit more the process do you still think there might be a laview drop out between the GPIB interface and HP?
I can now input data from Labview to HP Message 4 of Sorry, I’ve been on vacation. You can specify either a total sampling time llabview a specific number of points. Implementation of the addressing scheme is vendor specific and some vendors support mixed cases.
I have no experience with LAbView so far. Message 2 of Such error can occur if configuration of GPIB is not correct e. The driver uses Pascal calling conventions. What exactly is stopping us from doing this?
Message 4 of 6. Message 10 of The default error message for error 6 indicates an abort in the GPIB communication. I have a very old VI with a very 456b driver where you can obviously not choose an SMU or supply any source but can only read.